(U-Th)/He Gas Lab
UT (U-Th)/He Laboratory Picking and Grain Measurement Facilities
This facility is housed in a complete renovated laboratory space at the UT JSG. Mineral grains are handpicked and screened for inclusions using customized Nikon SMZ-U/100 stereomicroscopes with a rotating stages. The stereomicroscopes have both transmitted (polarized) and reflected light capabilities. Prior to loading samples into Pt sleeves, all grains are digitally photographed using a Nikon digital ColorView® camera and all digital pictures are archived. AnalySIS® imaging software is used to morphometrically measure each grain before loading into Pt tubes. These morphometric values are subsequently imported into a LabView routine to calculate the alpha-ejection correction. The (U-Th)/He picking and grain measurement facilities were funded through JSG start-up funds to Stockli and are also used for U-Pb (LA-ICP-MS and TIMS) and other geochronologic sample preparation.
Automated He Extraction Line for (U-Th)/He Dating
The UT (U-Th)/He laboratory houses three state-of-the-art, all metal, ultra-high vacuum noble gas extraction and purification lines for measuring 4He. The quadrupole He mass spectrometry systems consist of the following principle components: 1) One Photonmachine 30W Diode Laser and two Photonmachine 75W Diode lasers and a U.S. Laser continuous-mode Nd-YAG lasers for total fusion He laser extraction, ideal for single-crystal work (see House et al., 2000), 2) two separate all-metal extraction lines equipped with computer-controlled pneumatic Nupro valves and pumped by a combination of ion, turbo and rough pumps, 3) precise volume aliquot systems for spiking sample gas with 3He for isotope dilution, 4) precise volume aliquot systems for delivering a 4He standard with separate depletion tank systems to monitor 4He tank depletion, 5) gas purification system consisting of two SAES NP10 getters and two Janis cryogenic trap capable of separating He from other gases by variable temperature release at 16-37K, and 6) two Blazers Prisma QMS-200 quadrupole mass spectrometer for measuring 3He/4He ratios. Our helium isotope dilution procedure allows very low-blank (< 1 femtomole) and high-precision (<1-2%) measurements of 4He in dated crystals while allowing for recovery of the crystals for U and Th measurement in the same aliquot (see sections 5 and 6).
The UT extraction line components and valves are fully interfaced with a central computer and are fully automated using LabView software. The 3He spiking, cryogenic trap temperature cycling, and mass spectrometric analysis of samples, gas standards, and blanks are fully computer controlled. On extraction line #1, the laser heating of samples is controlled through a feedback loop using a video capture card allowing for continuous computerized adjustment of the laser output power for even heating. The laser sample planchet sits on a computer-controlled Newport X-Y stage and holds up to 44 samples. On laser extraction line #2, diode laser heating is controlled by an in-line two-color pyrometer. The 44 position planchet is fixed while the x-y-z position of the laser beam deliver is computer controlled.
Automated He Extraction Line for Diffusion Experiments
The UT (U-Th)/He laboratory also houses a third state-of-the-art, all metal, ultra-high vacuum noble gas extraction and purification line for measuring 4He dedicated to He diffusion experimental work and step-heating He extraction, but also has the potential for in-vacuum dissolution (acid digestion) degassing. The second quadrupole He mass spectrometry system consists of the following principle components: 1) four automated He diffusion experiment apparatuses (see details below), 2) an all-metal extraction line equipped with computer-controlled pneumatic Nupro valves and pumped by a combination of ion, turbo and rough pumps, 3) a precise volume aliquot system for spiking sample gas with 3He for isotope dilution, 4) a precise volume aliquot system for delivering a 4He standard with a separate system to monitor 4He tank depletion, 5) a gas purification system consisting of two SAES NP10 getters and a Janis cryogenic trap capable of separating He from other gases by variable temperature release at 16-37K, and 6) a Blazers Prisma QMS-200 quadrupole mass spectrometer for measuring 3He/4He ratios. Our helium isotope dilution procedure allows very low-blank (< 1 femtomole) and high-precision (<1-2%) measurements of 4He in analyzed crystals.
He Diffusion Experimental Line
As discussed above, it is essential to accurately quantify the helium diffusion characteristics of mineral phases in order to use them for thermochronological purposes. Diffusion properties such as activation energy and diffusivities (Do/a2) can be constrained by measuring fractional helium release in controlled step-heating experiments, which have traditionally been carried out using resistance furnaces. The UT (U-Th)/He laboratory is equipped with four specially designed diffusion cells to perform detailed in-vacuo step-heating experiments (Farley et al., 1999). Inside the diffusion cell the Cu-foil wrapped sample is suspended by a K or J-type thermocouple and is heated by a 120V halogen bulb projected through a sapphire window. Precise temperature control is accomplished through a feedback mechanism involving a Watlow thermal controller and a phase-angle-fired Eurotherm power supply. Estimated temperature stability during each step is better than ±1°C. The Watlow thermal controllers are interfaced with the lab computer, allowing the helium diffusion experiments to be executed in a fully automated mode using LabView software.
Most of our experimental efforts to date have concentrated on investigating the diffusion characteristics of a variety of new mineral phases, such as monazite, rutile, magnetite, fluorite, perovskite, or garnet. Extensive work on monazite has demonstrated a closure temperature of between 190 and 250°C (assuming a cooling rate of 10°C/m.y.) that is significantly dependent on composition and slightly dependent on grain size. The Arrhenius plot in Figure 3b shows an example of a monazite diffusion experiment using multiple heating cycles between 400 and 650°C.
Helix SFT Noble Gas Mass-Spec and Extraction Line for 4He/3He Analysis (coming in April 2012!!!)
Future expansion of the UT (U-Th)/He laboratory in 2012 will add a Thermo Helix SFT magnetic sector mass-spectrometer with dedicated UHV extraction and purification line. The line will be equipped with a diode laser for laser heating, two light-bulb furnaces for low-temperature step heating, and a PhotonMachine Analyte G.2 Excimer Laser Ablation system for laser-ablation He dating development. The Analyte G.2 will be share with one of the Thermo Element2 HR-ICP-MS instrument. Addition of this state-of-the-art instrumentation will allow the UT (U-Th)/He laboratory to engage in 4He/3He thermochronometry, analyze very young low He-yield volcanic samples and other low radiogenic samples (e.g., young magnetite), and also enter the world of He laser ablation measurements.
Pictures of June 2012 Installation!!!!!